188.8.131.52 NDI Equipment Settings.
Make the following initial settings on the Eddy Current Inspection Unit, NORTEC-1 9e".
- 200 KHz
Refer to Eddy Current Method, paragraph 1.4.11. Set up on test block as follows:
Null probe on test block.
Adjust phase as required to obtain horizontal lift-off.
Move probe over all three notches in test block. Adjust gain to obtain a three block vertical signal when
probe is passed over 0.040-inch notch in test block. (See the standard instrument display shown in Figure
184.108.40.206 Inspection Procedure. Refer to Eddy Current Method, paragraph 1.4.11 and Figure 6-13.
Place probe on a good area in the inspection location and null. Adjust phase as required to obtain horizontal lift-
Inspect the part.
Any signal similar to the notches in the test block are cause for rejection.
Either probe identified in paragraph 220.127.116.11 may be used depending primarily on the ease of
accessibility and user friendliness. If the probes are changed, steps 18.104.22.168 b. (1), (2), and (3)
shall be repeated each time a change is made.
22.214.171.124 Marking and Recording of Inspection Results. Mark and record inspection results as required by paragraph 1.3.
6.13.4 Backup Method. Fluorescent Penetrant, refer to paragraph 1.4.7.
6.13.5 System Securing. The pilot/copilot collective stick support, if removed, requires installation in accordance with
the applicable technical manuals listed in Table 1-1.