0019 00-1/(0019 00-2 blank)
UNIVERSAL TRACKING DEVICE (UTD)
INSPECTION AND FAULT ISOLATION
Workbench with ESD capability
TMDE Support Specialist
Tools and Equipment:
As listed in TM 1-6625-736-13&P, NSN 6625-01-347-8667 P/N 29085800
Operators Manual for Test Set, Electronic NSN 6625-01-347-8667
AVA system on workbench, DAU cover removed and connected to test set.
a. Examine UTD for visible damage. Replace items that are broken. See appropriate work packages for removal
and installation procedures.
b. Connect the DAU to the test set following instructions in the Operators Manual for Test Set, Electronic,
Section V-Calibration paragraph 2-11 (2) SETUP.
c. Perform Operators manual instructions Section V-Calibration, paragraph 2-11 (1) PROGRAM THE CADU, if
not previously accomplished.
d. Perform Operators manual instructions Section V-Calibration, paragraph 2-11 (5) UTD Test.
e. If no errors are reported, verify UTD values in Table 2-1, UTD Expected Output. If values are outside of
tolerance given, open UTD, inspect and replace PWA as outlined in WP 0051-001/2.
Repeat steps b. through f.
END OF WORK PACKAGE